The NSK VarioSurg 3 Control Unit, equipped with its specially-made engine, offers various advanced functions for ultrasonic bone surgery. It incorporates cutting-edge features that enhance cutting efficiency, precision, and safety. This intelligent power control system ensures precise and optimal performance during bone dissection. Another important function is the Burst Function, which adds a powerful vibration, often referred to as the “hammer effect,” to the vibrating tip. This feature enhances the ability to cut through hard tissue, making it easier and more efficient.
The system also prioritizes safety by minimizing the risk of damage to soft tissues, even in the event of accidental contact. Furthermore, the cavitation effect of the irrigation solution helps keep the surgical site blood-free, contributing to enhanced precision. The VarioSurg system’s handpiece is designed with the comfort of the dental professional in mind. It features a comfortable grip, superior ergonomics, and a slim and lightweight body, which minimizes hand fatigue and allows for greater accessibility to difficult oral regions. The integration of twin LEDs at the head of the handpiece ensures outstanding visibility, making treatments safer and more precise. Additionally, the advanced ultrasonics in the handpiece eliminates heat at the operation site, enabling minimally invasive procedures
The NSK LED handpiece used with the VarioSurg system is specifically designed to be shorter and lighter, achieving a perfect balance. This design improvement makes surgical procedures easier and stress-free, enabling dental professionals to provide more precise and efficient treatment.
- Ultrasonic bone surgery capabilities
- Feedback & Auto Tuning Function
- Burst Function for enhanced cutting
- Superior cutting power
- Minimally invasive bone dissection
- Safe for soft tissue protection
- The precision with cavitations effect
- Comfortable grip handpiece
- Slim and lightweight design
- Rated voltage: AC230V 50/60Hz
- Drive frequency: 28 – 32kHz
- Maximum output: 25W
- Power input: 54VA
- Dimension: W265 x D220 x H103mm
- Weight: 3kg
- Model: FC-78
- Cord Length: 2m
- Dimension: W268 x D230 x H103mm
- Weight: 1.4kg
- Model: VS3-LED-HPSC
- Cord Length: 2m
- Type of oscillation: Piezo type
- Optic: White LED
- Consumption current(LED): 0.17A (3.5V)
- Dimension: ø20×129mm (except cord)
- Weight: 70g (without cord)
- 1 x Foot Control
- 1 X Control Unit
- 1 X AC Power Cord
- 1 X LED handpiece with cord (Unshielded 2m)
- 5 X Irrigation Tube
- 1 X Irrigation Pole
- 1 X Handpiece stand
- 7 X Tube Holder
- 1 X Tip Wrench
- 6 X Tip
- 1 X Tip Holder
- 1 X Sterilization Cassette
Direction to use:
- Ensure that the device is connected to the appropriate power source,
- Familiarize yourself with the control panel and buttons of the VarioSurg 3. The key functions are as follows:
- LIGHT key: Used to select the Handpiece illumination intensity.
- COOLANT key: Used to select the Coolant Flow from 5 levels.
- P key: Used to set the PERIO Mode.
- E key: Used to set the ENDO Mode.
- S key: Used to set the SURG Mode.
- BURST key: Used to set the Burst Output Level (only for the SURG Mode).
- Turn on the device using the power switch, if applicable.
- Select the desired mode by pressing the corresponding key on the control panel. The available modes are SURG, ENDO, and PERIO.
- For the SURG mode, you can also activate the Burst function by pressing the BURST key. Adjust the Burst Output Level according to your preference.
- Adjust the power range for cutting by using the appropriate controls. In SURG mode, you can set the power range from 10% to 150% in multiples of 10%. In ENDO mode, the power range is from 5% to 100%, with increments below 50% in multiples of 5% and increments of 50% and larger in multiples of 10%.
- Set the Coolant Flow level by pressing the COOLANT key. Choose from the available 5 levels (OFF, LOW, HIGH) according to your requirements.
- Adjust the Handpiece illumination intensity using the LIGHT key. Select the desired brightness level for optimal visibility during the procedure.
- Once you have configured the desired settings, you can begin using the VarioSurg 3. Ensure that you have the appropriate handpiece attached for the specific procedure you are performing.
- During the operation, handle the handpiece with a comfortable grip. The VarioSurg 3 is designed with superior ergonomics to minimize hand fatigue and provide easy access to difficult oral regions.
- Pay attention to the operation site and adjust the power, coolant flow, and burst output level as needed for optimal results and safety